arXiv:gr-qc/0102045 11 Feb 2001

Angular Noise in Gravitational Wave Detectors

Gabriela González

Center for Gravitational Physics and Geometry The Pennsylvania State University 104 Davey Laboratory, University Park, PA 16802, USA

Angular fluctuations of suspended mirrors in gravitational wave interferometers are a source of noise both for the locking and the operation of the detectors. We describe here some of the sources of these fluctuations and methods for the estimation of their order of magnitude.

In interferometric gravitational wave detectors such as LIGO, the mirrors are suspended on fine wires. In designs for advanced detectors, the pendulums where the mirrors hang have multiple stages with both horizontal and vertical soft spring constants. The pendulum frequencies (around or below 1 Hz) are much lower than the frequency with optimal sensitivity for gravitational waves (around 100 Hz). The suspended configuration, which is necessary for many reasons (free masses, seismic isolation and immunity to thermal noise), has the disadvantage of allowing large pendular motions that need to be controlled to keep the detectors in a linear regime. Much attention has been paid to the horizontal motion of the masses, since they are directly observed by the interferometer and thus limit the sensitivity to gravitational waves.

However, the angular motion of the mirrors can also be an important source of sensitivity degradation, especially in long baseline optical cavities^1. Requirements for the acceptable amount of angular noise come in two flavors: (i) How much before we cannot lock easily the interferometer? (ii) How much can be tolerated while in operation? For the LIGO detectors now in construction, answers are (i) 0.5 mrad rms and (ii)10�^8 rad rms, and contributions less than 5% of shot noise^2. For advanced interferometers, the requirements may be up to 10 times tighter, which is very difficult to achieve: this highlights the need for their accurate estimation.

The identified sources of angular noise are straight seismic excitation (depends on pendulum mechanics); seismic excitation through mechanical cross-coupling; electronic excitation due to sensing/control noise; electronic excitation through actuators. cross couplings; and the always unavoidable thermal noise.

Seismic noise produces angular noise through straightforward pendulum dynamics. In LIGO these calculations predict motions smaller than the required tolerance. However, due to mechanical asymmetries, it is likely that a larger angular motion is produced by cross couplings of other degrees of freedom. This problem will be compounded in multiple pendulums, and needs to be carefully studied. We have an experimental program in Penn State to such effect^3.

The control of the masses done to damp the pendular motions is usually applied as several forces on different points of the mirror that add up to the required forces and torques derived in a feedback mechanism from sensors at the same points where the actuation is done. The sensed signals can be decomposed in the natural suspension modes, with a different feedback mechanism for each mode (“modal damping”), as is done on LIGO suspensions, or they can be sensed and acted upon independently, as long as the system is not overdetermined (“point-to-point damping”), as is done in GEO600 suspensions. Both of these systems, however, suffer the risk of introducing excess forces due to the imperfect and unbalanced sensing or actuation. The sensor noise (ß 10�^9rad=pHz) introduces excess angular motion just below the tolerances for the LIGO detector, but the same sensors will be too noisy for advanced LIGO detectors. This is however, a problem only relevant for locking acquisition: when in operation, interferometric (“wavefront”) sensors can be used whose noise is negligible compared with tolerances. The limiting noise in the actuators is severely constrained by the maximum horizontal motion they can produce, so there are no additional constrained posed by the angular noise they may produce. However, the angular noise introduced by unbalanced forces is a serious problem in the LIGO detectors that needs a careful tuning of the actuators, taking into account the pendulum dynamics^4.

The power spectral density of the thermal noise, or brownian motion, of the mirrors is determined by the dissipation mechanisms in the suspension, which in the ideal situation are due to material properties of the suspensions wires. A calculation of the angular thermal motion can be done using the Fluctuation-Dissipation theorem, but the needed calculation is the observation of the angular motion by the interferometer: this depends on the angles between the axes of the optical cavities and the suspension axes. This calculation has been done for LIGO suspensions by the author^5, and shows that the angular thermal noise will not limit the detector sensitivity. However, this calculation needs to be done for advanced detectors, where angular noise may represent a significant contributor.

In conclusion, we suggest that angular noise of suspended mirrors does not limit the sensitivity of LIGO detectors, but only when extreme care is taken care with control cross couplings. The safety margin is less than an order of magnitude for some of the sources, which suggests that angular noise needs to be carefully studied in advanced LIGO configurations.

References

1. Alignment of an interferometric gravitational wave detector, P Fritschel, G González, N Mavalvala, D Shoemaker, D Sigg, M Zucker, Applied Optics 37(28),6734 (1 October 1998) 2. Alignment Sensing/Control (ASC) Design Requirements Document, Peter Fritschel, LIGO Internal Document T952007-04-I,1997.

3. Development of a Double Pendulum for Gravitational Wave Detectors, M.

Beilby, G. González, M. Duffy, A. Stuver, J. Poker, in GRAVITATIONAL WAVES: Third Edoardo Amaldi Conference, S. Meshkov, ed., California Institute of Technology, Pasadena, California, AIP Conference Proceedings 523, June 2000 (gr-qc/9911027).

4. Suspension Controller Tuning Procedure, Mark Barton Gabriela González, Steve Penn, LIGO Internal Document T000003-A-D (September 12 2000).

5. Suspensions thermal noise in the LIGO gravitational wave detector, Gabriela

González, Classical and Quantum Gravity 17(21),4409 (7 November 2000)