arXiv:hep-ex/0305074 v1 26 May 2003

The 28th International Cosmic Ray Conference 1

The AMS-02 Time of Flight System. Final Design

Diego Casadei,^1 on behalf of the AMS TOF Collaboration: V. Bindi,^1 N. Carota,^1 D. Casadei,^1 G. Castellini^2, F. Cindolo,^1 A. Contin,^1 F. Giovacchini,^1 P. Giusti,^1 G. Laurenti,^1 G. Levi,^1 R. Martelli,^1 F. Palmonari,^1 L. Quadrani,^1 M. Salvadore,^1 C. Sbarra,^1 and A. Zichichi^1 (1) Bologna University and INFN Bologna, via Irnerio 46, I-40126 Bologna, Italy (2) CNR-IFAC, Via Panciatichi 64, I-50127 Firenze, Italy

Abstract

The AMS-02 detector is a superconducting magnetic spectrometer that will operate on the International Space Station. The time of flight (TOF) system of AMS-02 is composed by four scintillator planes with 8, 8, 10, 8 counters each, read at both ends by a total of 144 phototubes. This paper describes the new design, the expected performances, and shows preliminary results of the ion beam test carried on at CERN on October 2002.

1. Introduction

The Alpha Magnetic Spectrometer (AMS) is a particle detector that will be installed on the International Space Station (ISS) in 2005 (NASA shuttle flight UF-4.1) to measure Cosmic Ray (CR) fluxes for at least three years.

During the precursor flight aboard of the shuttle Discovery (NASA STS-91 mission, 2–12 June 1998), the test detector AMS-01 was operated for about 180 hours, collecting over one hundred millions CR events [1]. The AMS-02 detector, that will operate on ISS, is described in ref. [2]. Scientific goals of AMS-02 are:

• improved measurement of the antimatter fraction in cosmic rays;

• search for exotic particle annihilation signatures in the proton, electron and γ-ray spectra over the energy range 1–10^3 GeV;

• high statistics measurements of the CR ion spectra below 1 TeV per nucleon.

2. The time of flight system

The time of flight (TOF) system of AMS-02, developed in the laboratories of INFN Bologna, will be somewhat different from the TOF system of AMS-01 [3]. The whole system still consists of four plastic scintillator planes (two above

pp. 1–4 © 2003 by Universal Academy Press, Inc.

Fig. 1. The scintillator paddles of the AMS-02 TOF system without light guides and phototubes.

Fig. 2. A complete TOF counter with 2 + 2 PMT and curved light guides.

and two below the magnet), but with 8, 8, 10, 8 counters each (instead of 14), as shown in figure 1. The counters of adjacent planes are orthogonal as in AMS-01, in order to provide a certain granularity at the trigger level, and in each plane they are overlapped by 5 mm. Howewer, the outermost counters of AMS-02 have a trapezoidal shape in order to reduce the weight.

All 1 cm thick scintillators are read at both ends by independently powered photo-multiplier tubes (PMT), connected with transparent light guides, as shown in figure 2. Due to the strong residual magnetic field in the PMT zones, the AMS- 02 TOF counters adopt the Hamamatsu R5946 phototubes [4] and have curved light guides. The total number of PMT is 144 and the optical contact with the light guides has been realized using soft transparent pads, that also provide a good mechanical coupling. The same method was adopted for AMS-01 and proved to be able to survive both to the launch and to the landing of the shuttle Discovery without any damage to the PMT.

Fig. 3. Charge peaks observed with the C1 scintillator, using left and right anodes (“C1LA” and “C1RA”, left panel) and dynodes (“C1LD” and “C1RD”right panel, where the first peak is He). The 3 cm wide beam crossed the counter in the center and the gain ratio of the two sides was 3:2.

The TOF system has the following essential tasks. First, its signals are used to form the fast trigger (the very first level of the data acquisition chain). Second, the particle time of flight is used to measured the velocity β = v/c and to distinguish between upward and downward going particles. This is fundamental in order to separate particles from antiparticles. Third, the energy loss measurement is used by the TOF system to send to the trigger box a special flag for ions events, and to provide an independent particle charge measurement.

3. The ion beam test preliminary results

Two AMS-02 TOF counters with 1×12×130 cm^3 polyvinyltoluene scintillators (Eljen EJ-200, “C1”, and Bicron BC408, “C2”), together with one AMS-01 counter used as reference, were tested at CERN on October 2002 with standard NIM and CAMAC electronics, on the ion beam provided by the SPS facility. The primary Pb beam was directed against 10–30 cm long Be targets, producing secondary particles and nuclei with charge spanning a very wide range: Z = 1–82. The H8 selection line was tuned to obtain secondaries with A/Z = 2, 3/2, 7/4 and 1. For what concerns the scintillators, all primary particles were at their minimum ionization plateau (that is at Lorenz factors > 3).

The square root of the anode and dynode signals (in arbitrary units) of the first scintillator during runs 105–114 is shown in figure 3. The dynode peaks

Fig. 4. Birks’ law fit of the dynode peaks (left panel), and average time of flight resolution as function of the charge (right panel).

were fitted with gaussians and then the second order Birk’s law: Q = A^{dE}_{dx}

1 + B^{dE}_{dx} + C (_{dE} dx )_2 =

p_1Z^2 1 + p_2Z^2 + p_3Z^4

(where the constants A, B, C are “adsorbed” by p_1, p_2, p_3) was used to obtain the best estimate of the particle atomic number Z from the measured charge Q = ADC − p_4 (p_4 is the fitted pedestal position), as shown in the left panel of figure 4. The combined TOF and RICH charge measurement is shown in ref. [5].

Finally, the resolution of the measured time of flight between each pair of counters can be evaluated as function of the particle charge, as shown in the right panel of figure 4. The time resolution improves with increasing charge due to the higher photostatistics, until it reaches a level dominated by the electronic noise [3]. With the standard electronics used at CERN, this level was 57 ± 1 ps, while the maximum jitter allowed for the AMS-02 TOF front-end boards is 50 ps.

We wish to thank the organizations and individuals acknowledged in ref. [2].

4. References

1. Aguilar M. et al., Physics Reports, vol. 366/6 (2002), 331-404; 2. Gentile S., “The Alpha Magnetic Spectrometer on the International Space Station”, these proceedings;

3. Alvisi D. et al., NIM A 437 (1999) 212-221;

4. Brocco L. et al., Proc. of the 27th ICRC (2001), 2193-2196; 5. Buénerd M., “The AMS RICH detector”, these proceedings.